No products in the cart.
Roll over image to zoom in
Click to open expanded view
Ormco Enlight Light Cure Adhesive Introductory Kit
Ormco Enlight Light Cure Adhesive
Enlight Bonding System
Highly filled light-cure adhesive. Bis-GMA resin for superb handling characteristics, easy cleaning.
Highly active initiater system supports reduced cure times.
- Place the lip/cheek retractor and saliva ejector in the mouth. prphy the teeth thoroughly with pumice and water. Rinse with water while using a high speed evacuation system. Dry the teeth with air syringe.
- Apply the etchant solution to the bonding surfaces with applicator sponges. Gently dab the etching solution : do no rub. A continuous dabbing of fresh solution for 30 secdonds is most effective. Rinse thoroughly with a forcefull air-water spray and suction using high speed evacuator. Maintain a dry field. Do no allowthe patient to rinse or permit contact of the etched enamel with saliva or mucosal tissue.
- Dry the etched enamel with clean, dry air, The etched enamel must now have a uniform, dull frostly appearance.
- Dispense a small amount of sealant on the mixing pad. apply ortho solo sealant to teeth using the brushes provided. A thin, uniform coating should extent from the gingival crest and vover all of the etched enamel. Replance the cap immediately after dispensing sealant.
- Extrude a small amount of enlight adhesive paste into the base of the bracket pad. After application, clean the tip and replace cap.
- Place the bracket onto the tooth surface immediately after applijng adhesive.\
- Adjust the bracket to its final position and press firmly.
- Remove the excess adhesive from the periphery of the brackets base.
- With Demetron’s optilux 501, Expose the adhesive for 10 second for metal brackets, and 5 second for ceramic brackets. shining the light directly through the bracket bonds ceramic brackets. Divide the curing time and shine the light on the mesial and distal edges of the brackets to bond metal brackets.
- Active archwires may be placed immediately after curring the last bracket.
Shelf life and storage
- Do not expose materials to intense light or elevated temperatures.
- Enlight should be used at ambient temperature.
- Etching Solution :Contains 37% phosphoric acid solution. Avoid contact with skin, eyes, and soft tissue. In case of contact with skin or eyes, flush immediately with water, get medical attention for eyes. Do not take internally.
- Sealant and paste : Avoid prolonged or repeated contact with oral tissues and skin. wash skin with soap and water after contact. uncured materials may cause contact dermatitis. if skin irritation occurs, discontinues use. Use in a well ventilated area. Keep containers closed when not in use.